Buy Surgical Instrument Washers that are manufactured in America.
Surgical Instrument Washers built with locally available nonproprietary replacement parts.
Product Support: 1-509-747-5027
The Surgical Instrument Washer is a heavy-duty, high volume, single chamber, cabinet style washer designed for thorough and efficient cleaning of laboratory glassware, utensils, and miscellaneous items used in research and healthcare facilities.
Surgical Instrument Washer SIZE:
Compartment usable space: 26” wide x 26" high x 29" deep
Overall Unit”: 42" wide x 80" high x 36" deep Sliding Door(s)
The Surgical Instrument Washer shall be provided with a combination of rotating jet spray and direct spindle spraying for all treatment solutions. The Surgical Instrument Washer spray system consists of a single rotating spray arm mounted above, and interchangeable spindle wash or a rotating jet spray arms mounted below for internal treatment of the racked items. The bottom header automatically couples to the re-circulated wash/rinse treatment system. Surgical Instrument Washer door(s) shall be provided with a full view, tempered, double glass viewing window. Surgical Instrument Washer OPERATION. The operator loads the items to be cleaned in the wash chamber, closes the door, selects the proper cycle, and depresses the Cycle Start button. The washer proceeds through the treatment schedule and automatically shuts off at the completion of the cycle. The operator then opens door and removes the cleaned items. Surgical Instrument Washer PRE-WASH. Either Hot water from the house supply fills the sump, or Water remaining in the chamber sump from final rinse of the previous cycle is re-circulated through the jet system under pump pressure and pumped to drain upon completion. Phase is adjustable from 0 – 9999 seconds in 1-second intervals and 120F - 190F intervals. Surgical Instrument Washer DETERGENT WASH. Hot water from the house supply fills the sump and is pumped through the jet system. As required, a specified amount of detergent is added during recirculation using the OPTIONAL supplied, or customer supplied detergent injection pumps. At the end of the treatment, the detergent solution is pumped to drain. Phase is adjustable from 0 – 9999 seconds in 1-second intervals and 120F - 190F in 1 intervals. Surgical Instrument Washer RINSE.
Hot water from house supply fills the chamber sump and is re-circulated through the jet system under pump pressure to remove residual detergent. Re-circulated water is pumped to drain on completion of treatment. Surgical Instrument Washer DETERGENT WASH II. Same as Detergent Wash phase – allowing a secondary detergent to be utilized.Surgical Instrument Washer PURIFIED WATER RINSE(S)
Hot water from house supply fills the chamber sump and is re-circulated through the jet system under pump pressure to remove residual detergent. Re-circulated water is pumped to drain on completion of treatment.